Spine imaging after lumbar disc replacement: pitfalls and current recommendations
© Robinson and Sandén; licensee BioMed Central Ltd. 2009
Received: 09 April 2009
Accepted: 20 July 2009
Published: 20 July 2009
Most lumbar artificial discs are still composed of stainless steel alloys, which prevents adequate postoperative diagnostic imaging of the operated region when using magnetic resonance imaging (MRI). Thus patients with postoperative radicular symptoms or claudication after stainless steel implants often require alternative diagnostic procedures.
Possible complications of lumbar total disc replacement (TDR) are reviewed from the available literature and imaging recommendations given with regard to implant type. Two illustrative cases are presented in figures.
Access-related complications, infections, implant wear, loosening or fracture, polyethylene inlay dislodgement, facet joint hypertrophy, central stenosis, and ankylosis of the operated segment can be visualised both in titanium and stainless steel implants, but require different imaging modalities due to magnetic artifacts in MRI.
Alternative radiographic procedures should be considered when evaluating patients following TDR. Postoperative complications following lumbar TDR including spinal stenosis causing radiculopathy and implant loosening can be visualised by myelography and radionucleotide techniques as an adjunct to plain film radiographs. Even in the presence of massive stainless steel TDR implants lumbar radicular stenosis and implant loosening can be visualised if myelography and radionuclide techniques are applied.
Alloy and articulation specifics of available lumbar TDR implants
Non-constrained PE inlay
It is widely accepted that lumbar disc arthroplasty is associated with several access- and implant-associated complications. Unfortunately imaging modalities after TDR cannot simply be transferred from fusion patients. In this review the complications following total disc replacement are discussed and guidelines for imaging following lumbar disc arthroplasty are provided.
Imaging modalities after lumbar total disc replacement
Imaging guidelines after lumbar total disc replacement with regard to implant alloy
Stainless steel (CoCr)
Metallosis/PE wear loosening
Plain x-ray, scintigraphy, PET-CT
Plain x-ray, scintigraphy, PET-CT
Plain x-ray, CT
Plain x-ray, CT
Plain x-ray, CT
Plain x-ray, CT
MR, PET-CT, radiolabelled white blood cell scintigraphy
PET-CT, radiolabelled white blood cell scintigraphy
Late facet degeneration/secondary stenosis
Plain x-ray, CT
Plain x-ray, CT
Plain x-ray, CT
Plain x-ray, CT
CT, dynamic plain x-ray
CT, dynamic plain x-ray
Myelography alone and later combined with CT has for many years been the primary diagnostic means in suspected lumbar spinal stenosis. Myelographic findings in central canal stenosis include complete or partial block to the contrast column, appearing as an hour-glass constriction on the AP view. There appears to be poor correlation between clinical symptoms and findings at myelography, with myelography commonly showing more abnormalities than would be expected [6, 7], although some authors found a block to the flow of contrast medium to be a good predictor of successful outcome following spinal decompression .
Scintigraphy with 99mTc radionuclide tagged white blood cells can be a helpful means in diagnosing suspicious infections . 99mTc-hexamethylpropylene amine oxime (HMPAO) is generally more sensitive for the detection of acute osteomyelitis than of chronic osteomyelitis (with a high sensitivity 97.7% and a specificity of 96.8%) . 99mTc-HMPAO is also able to separate septic from aseptic bone lesions . Leukocyte imaging can be applied to chronic osteomyelitis (including infected joint prostheses). 111In-Oxine is preferred over 99mTc-HMPAO in such a case because it can be performed simultaneously with 99mTc-stannous fluoride colloid scanning for the assessment of bone marrow involvement (mismatched defects identify infectious foci). Infected orthopaedic hardware will have increased uptake of 111In-oxine, whereas activity will be normal or decreased in healthy or loose prostheses .
Postoperative low-grade-infection or implant loosening can be hard to visualise. As in hip or knee arthroplasty radionuclide scintigraphy has some value with regard to sensitivity, but has lower specificity with regard to loosening or low-grade infection. Gallium 67 scans have a sensitivity and specificity of 89% and 85% .
In the last years the increased availability of positron emission tomography (PET) enabled postoperative diagnostics with greater specificity for infection. Gratz et al  found fluorine-18 fluorodeoxyglucose (FDG) hybrid PET superior to MRI, 67Ga citrate and 99mTc-methylene diphosphonate (MDP), especially in patients with low-grade spondylitis, adjacent soft tissue infections and advanced bone degeneration. To visualise increased bone turnover 18F-flouride-PET is an option, and found helpful to diagnose aseptic artificial knee joint loosening . There is no clinical evidence available with regard to PET after TDR, but is very likely that this imaging modality will emerge to a more specific diagnostic tool and replace 99mTc-MDP scintigraphy in cases with suspected infections or implant loosening where MRI is obsolete .
Indication and options for imaging after lumbar total disc replacement
Despite the biomechanically appealing concept of motion preservation, lumbar disc arthroplasty is associated with access-related and implant-specific complications. Access and implant related complications can be due to technical errors or may be due to abnormal anatomy . Implant wear and loosening are known to occur in all types of arthroplasty.
a) Access-related complications
All commercially available FDA-approved lumbar total disc replacements require an anterior access for the implantation of artificial discs. Lumbar discs can be implanted via a transperitoneal or a retroperitoneal approach, of which the latter is favoured by most surgeons. Complications with regard to these accesses are well known and differ not from those of anterior fusion, even though TDR-implantation requires a wider exposure of the disc, which often requires more traction and pressure of retractors on the surrounding tissue . Especially revision-surgery is associated with higher morbidity and mortality compared to the index surgery . Revision may require a transperitoneal approach if the index intervention was retroperitoneal. Then vascular and ureteric stenting may reduce the risk for access-related complications. Nevertheless the availability of a vascular surgeon on call is highly recommended in revision cases .
Most access-related complications can be visualised by sonography and abdominal CT with intravenous contrast. In few cases an angiography or an uretherography may be necessary.
There is no specific data on the occurrence of infection of artificial discs, but early and mid-term infections may occur as likely as in anterior fusion with 0 and 12% in lumbar instrumented fusion . Nevertheless due to the prevalence of motion and a neo-joint formation, late haematogenic spread infection as known from hip and knee arthoplasty is possible in the immunocompromised TDR patient, with detrimental effects on the patient's health and function.
Postoperative infections of the spine can optimally be visualised by MR with contrast. A sensitivity and specificity of 93% and 97% makes the MRI superior to all other radiological imaging modalities . If MRI-scans are not available or stainless steel implants were used radionuclide imaging should be used to reveal focal increased uptake. If available PET-CT scans should be considered to diagnose a suspicious postoperative spinal infection, since PET has a higher specificity than radionuclide scintigraphy.
c) Implant dislocation and polyethylene inlay dislodgment
Mobile polyethylene cores can dislocate, too, and can cause compression to anterior abdominal or pelvic structures. This is especially true in non-constrained PE inlays (i.e. Charité TDR) Mathew et al  presented a case of an anterior polyethylene dislocation after lumbar TDR (Charité), which was then revised and fused with an anterior standalone cage. Within the case series by Kurtz et al  were 2 further cases and in the series of Leary et al  one case of anterior core dislocations. This complication required in all cases anterior PE core extraction and fusion. Reposition or exchange of the PE core alone is not recommended due to the prevalent biomechanical failure of the implant with risk of recurrence of dislocation.
Implant dislocation can easily be diagnosed with plain radiographs (figure 1). This is also true for PE core dislocations, since the PE-core contains dense markers in the available lumbar implants.
d) Implant failure/fracture
The long-term investigation of the first implanted lumbar discs (n = 53, mean follow-up 17.3 years; Charité, Link) by Putzier et al  included 7 implant fractures resulting in segmental fusion, all of which occurred with the earlier Charité II-model. Of these only one case underwent surgical revision requiring secondary fusion due to intractable pain. These cases show that the chosen implant has to stand a continuous load, requiring enduring metal alloys and stable implant design to provide a lasting product.
Implant fractures are mostly visible on plain radiographs.
e) Implant loosening/wear
Since implant wear has for decades been a major issue in hip and knee-arthroplasty it is not surprising that other weight-bearing implants with articulating surfaces – as in lumbar disc prostheses – will have to deal with implant wear and loosening, too. Punt et al  retrieved 16 artificial discs (SB Charité III, DePuy) which where explanted due to intractable pain after a mean of 8 years. They found polyethylene (PE)-wear debris causing chronic inflammation due to mononuclear macrophage and giant-cell activation. Kurtz et al  investigated 38 artificial discs (SB Charité III, DePuy) that were explanted in patients with therapy-resistant chronic pain after an average of 7.3 years (range 2.0–16.1). They found a positive correlation between PE inlay dome penetration and implantation time (r = 0.46, p = 0.004) and an average penetration rate of 0.5 mm/year (range 0.01–0.18 mm/year). These results could be validated in a PE-wear in vitro model .
Metal-metal articulations are known to cause metallosis with chronic inflammation leading to implant loosening . François et al  presented a case of a loosened lumbar disc prosthesis with metal-metal combination (Maverick, Medtronic) one year after implantation causing persistent pain. They found in situ a loosening of the superior endplate and histologically typical signs of metallosis in the surrounding scar tissue. Cavanaugh et al  found similar hypergranulation patterns without histological metallosis surrounding a cervical explanted artificial disc causing re-stenosis 6 months after implantation. Implant loosening due to PE or metal wear is a known problem in TDR. Unfortunately not only loosening, but also symptomatic re-stenosis due to granulation caused by PE debris or metallosis were found.
Diagnostics of implant loosening range from standard plain radiographs – revealing radiolucencies around the implant – to scintigraphy and PET, showing increased metabolism in the bone-implant interface.
f) Central or lateral stenosis due to facet degeneration
MR diagnostics are the gold-standard for investigation of spinal canal pathologies. In cases where this is not possible – i.e. stainless steel TDR – other modalities as CT and myelography have to be considered.
g) Implant malpositioning
Bendo et al  found the midline retroperitoneal access associated with less sagittal implant misplacements than the pararectal lateral retroperitoneal access (p = 0.021), but this did not have any impact on clinical functional outcome, determined with the Oswestry disability index (p = 0.92). Rauschmann et al  presented in a cadaver study improved positioning of the lumbar artificial disc if navigation was applied compared to fluoroscopy (n = 30). Malpositioning can be due to minimal malrotation of the image intensifier, anomalities in the spinal process anatomy, and the parallax effect in intraoperative fluoroscopy . Navigation can therefore have significant impact on the quality and durability of artificial discs in the future.
To control implant positioning plain radiographs are usually sufficient.
h) Segmental ankylosis
Long-term investigations failed to present preservation of motion of the operated segment. Putzier et al  found that 60% of all treated patients (n = 53) had spontaneous fusion of the treated segment after a mean follow-up of 17-years (Charité, Link). Interestingly these patients were better than those with remaining motion with regard to functional measures (ODI, p < 0.05) and pain (VAS, p < 0.05). Thus fusion should be viewed as a natural course of motion preservation treatment and not as a complication.
To diagnose fusion either a CT-scan showing ankylosis or dynamic radiographs excluding remaining segmental mobility are required.
i) Progressive implant subsidence
In osteoporosis or poor bone quality, years after the original implantation, the artificial disc can subside into the endplates, causing secondary implant dislocation and increasing pain. To avoid this complication the choice of the largest possible implant and a placement close to the rim of the endplates is recommended .
To visualise implant subsidence plain radiographs are sufficient. If an osteoporotic vertebral endplate fracture is suspected, radionuclide scans may show increased uptake of the fractured vertebra.
Postoperative imaging of lumbar total disc replacement requires alternative imaging techniques to conventional MRI-diagnostics. All currently available lumbar artificial discs contain stainless steel alloys, and, therefore, generate major artefacts on MRI. Alternative imaging modalities include plain film radiographs, myelography and radionucleotide imaging. These imaging modalities should be utilised in cases with suspected postoperative complications after lumbar TDR.
The appearance of artificial discs on the implant market meant a drawback in postoperative radiological transparency with regard to MR-diagnostics, since all lumbar disc prostheses are still based on stainless steel alloys, thus causing major artefacts. Other imaging modalities are available and are in the first line of choice in cases with suspected postoperative complications after lumbar TDR. Until lumbar titanium artificial discs are available postoperative imaging of the operated segment requires altered radiological strategies.
Written informed consent was obtained from the patients presented in this article and is available to the editor-in-chief by request.
- Bono CM, Garfin SR: History and evolution of disc replacement. Spine J. 2004, 4 (6 Suppl): 145S-150S. 10.1016/j.spinee.2004.07.005.View ArticlePubMedGoogle Scholar
- Gibson JN, Waddell G: Surgery for degenerative lumbar spondylosis. Cochrane. Database Syst Rev. 2005, 4: CD001352-PubMedGoogle Scholar
- Sekhon LH, Duggal N, Lynch JJ, Haid RW, Heller JG, Riew KD, Seex K, Anderson PA: Magnetic resonance imaging clarity of the Bryan, Prodisc-C, Prestige LP, and PCM cervical arthroplasty devices. Spine. 2007, 32: 673-80. 10.1097/01.brs.0000257547.17822.14.View ArticlePubMedGoogle Scholar
- Matsuura H, Inoue T, Konno H, Sasaki M, Ogasawara K, Ogawa A: Quantification of susceptibility artifacts produced on high-field magnetic resonance images by various biomaterials used for neurosurgical implants. Technical note. J Neurosurg. 2002, 97: 1472-5. 10.3171/jns.2002.97.6.1472.View ArticlePubMedGoogle Scholar
- Suh JS, Jeong EK, Shin KH, Cho JH, Na JB, Kim DH, Han CD: Minimizing artifacts caused by metallic implants at MR imaging: experimental and clinical studies. AJR Am J Roentgenol. 1998, 171: 1207-13.View ArticlePubMedGoogle Scholar
- Amundsen T, Weber H, Lilleas F, Nordal HJ, Abdelnoor HJ, Magnaes B: Lumbar spinal stenosis. Clinical and radiologic features. Spine. 1995, 20: 1178-86. 10.1097/00007632-199505150-00013.View ArticlePubMedGoogle Scholar
- Jonsson B, Annertz M, Sjöberg C, Strömqvist B: A prospective and consecutive study of surgically treated lumbar spinal stenosis. Part I: Clinical features related to radiographic findings. Spine. 1997, 22: 2932-7. 10.1097/00007632-199712150-00016.View ArticlePubMedGoogle Scholar
- Herno A, Airaksinen O, Saari T, Miettinen H: The predictive value of preoperative myelography in lumbar spinal stenosis. Spine. 1994, 19: 1335-8. 10.1097/00007632-199409000-00019.View ArticlePubMedGoogle Scholar
- Seabold JE, Nepola JV: Imaging techniques for evaluation of postoperative orthopedic infections. Q J Nucl Med. 1999, 43: 21-8.PubMedGoogle Scholar
- Wolf G, Aigner RM, Schwarz T: Diagnosis of bone infection using 99mTc HMPAO labelled leukocytes. Nucl Med Commun. 2001, 22: 1201-1206. 10.1097/00006231-200111000-00006.View ArticlePubMedGoogle Scholar
- Datz FL, Seabold JE, Brown ML, Forstrom LA, Greenspan BS, McAfee JG, Palestro CJ, Schauwecker DS, Royal HD: Procedure guideline for technetium-99m-HMPAO-labeled leukocyte scintigraphy for suspected infection/inflammation. Society of Nuclear Medicine. J Nucl Med. 1997, 38: 987-90.PubMedGoogle Scholar
- Bruschwein DA, Brown ML, McLeod RA: Gallium scintigraphy in the evaluation of disk-space infections: concise communication. J Nucl Med. 1980, 21: 925-7.PubMedGoogle Scholar
- Gratz S, Rennen HJ, Boerman OC, Oyen WJ, Mast P, Behr TM, Corstens FH: 99mTc-HMPAO-labeled autologous versus heterologous leukocytes for imaging infection. J Nucl Med. 2002, 43: 918-24.PubMedGoogle Scholar
- David T: Revision of a Charité artificial disc 9.5 years in vivo to a new Charité artificial disc: case report and explant analysis. Eur Spine J. 2005, 14: 507-11. 10.1007/s00586-004-0842-x. Sterner T, Pink R, Freudenberg L, Jentzen T, Quitmann H, Bockisch A, Löer F: The role of [18F]fluoride positron emission tomography in the early detection of aseptic loosening of total knee arthroplasty. Int J Surg 2007, 5:99–104.PubMed CentralView ArticlePubMedGoogle Scholar
- van Ooij A, Kurtz SM, Stessels F, Noten H, van Rhijn L: Polyethylene wear debris and long-term clinical failure of the Charité disc prosthesis: a study of 4 patients. Spine. 2007, 32: 223-9. 10.1097/01.brs.0000251370.56327.c6. Zhuang H, Yang H, Alavi A: Critical role of 18F-labeled fluorodeoxyglucose PET in the management of patients with arthroplasty. Radiol Clin North Am 2007, 45:711–8, vii.View ArticlePubMedGoogle Scholar
- Leary SP, Regan JJ, Lanman TH, Wagner WH: Revision and explantation strategies involving the CHARITE lumbar artificial disc replacement. Spine. 2007, 32: 1001-11. 10.1097/01.brs.0000260794.73938.93.View ArticlePubMedGoogle Scholar
- Tropiano P, Huang RC, Girardi FP, Cammisa FP, Marnay T: Lumbar total disc replacement. Seven to eleven-year follow-up. J Bone Joint Surg Am. 2005, 87: 490-6. 10.2106/JBJS.C.01345.View ArticlePubMedGoogle Scholar
- Brau SA, Delamarter RB, Kropf MA, Watkins RG, Williams LA, Schiffman ML, Bae HW: Access strategies for revision in anterior lumbar surgery. Spine. 2008, 33: 1662-7. 10.1097/BRS.0b013e31817bb970.View ArticlePubMedGoogle Scholar
- Wagner WH, Regan JJ, Leary SP, Lanman TH, Johnson JP, Rao RK, Cossman DV: Access strategies for revision or explantation of the Charité lumbar artificial disc replacement. J Vasc Surg. 2006, 44: 1266-72. 10.1016/j.jvs.2006.07.046.View ArticlePubMedGoogle Scholar
- Beiner JM, Grauer J, Kwon BK, Vaccaro AR: Postoperative wound infections of the spine. Neurosurg Focus. 2003, 15: E14-10.3171/foc.2003.15.3.14.View ArticlePubMedGoogle Scholar
- Szypryt EP, Hardy JG, Hinton CE, Worthington BS, Mulholland RC: A comparison between magnetic resonance imaging and scintigraphic bone imaging in the diagnosis of disc space infection in an animal model. Spine. 1988, 13: 1042-8. 10.1097/00007632-198809000-00012.View ArticlePubMedGoogle Scholar
- Mathew P, Blackman M, Redla S, Hussein AA: Bilateral pedicle fractures following anterior dislocation of the polyethylene inlay of a ProDisc artificial disc replacement: a case report of an unusual complication. Spine. 2005, 30: E311-4. 10.1097/01.brs.0000164135.03844.b6.View ArticlePubMedGoogle Scholar
- Kurtz SM, Peloza J, Siskey R, Villarraga ML: Analysis of a retrieved polyethylene total disc replacement component. Spine J. 2005, 5: 344-50. 10.1016/j.spinee.2004.11.011.View ArticlePubMedGoogle Scholar
- Putzier M, Funk JF, Schneider SV, Gross C, Tohtz SW, Khodadadyan-Klostermann C, Perka C, Kandziora F: Charité total disc replacement – clinical and radiographical results after an average follow-up of 17 years. Eur Spine J. 2006, 15: 183-95. 10.1007/s00586-005-1022-3.PubMed CentralView ArticlePubMedGoogle Scholar
- Punt IM, Visser VM, van Rhijn LW, Kurtz SM, Antonis J, Schurink GW, van Ooij A: Complications and reoperations of the SB Charité lumbar disc prosthesis: experience in 75 patients. Eur Spine J. 2008, 17: 36-43. 10.1007/s00586-007-0506-8.PubMed CentralView ArticlePubMedGoogle Scholar
- Kurtz SM, Patwardhan A, MacDonald D, Ciccarelli L, van Ooij A, Lorenz M, Zindrick M, O'Leary P, Isaza J, Ross R: What is the correlation of in vivo wear and damage patterns with in vitro TDR motion response?. Spine. 2008, 33: 481-9. 10.1097/BRS.0b013e318165e3be.PubMed CentralView ArticlePubMedGoogle Scholar
- Korovessis P, Petsinis G, Repanti M, Repantis T: Metallosis After Contemporary Metal-on-Metal Total Hip Arthroplasty. Five to Nine-Year Follow-up. J Bone Joint Surg Am. 2006, 88: 1183-1191. 10.2106/JBJS.D.02916.View ArticlePubMedGoogle Scholar
- François J, Coessens R, Lauweryns P: Early removal of a Maverick disc prosthesis: surgical findings and morphological changes. Acta Orthop Belg. 2007, 73: 122-7.PubMedGoogle Scholar
- Cavanaugh DA, Nunley PD, Kerr EJ, Werner DJ, Jawahar A: Delayed hyper-reactivity to metal ions after cervical disc arthroplasty: a case report and literature review. Spine. 34: E262-5. 10.1097/BRS.0b013e318195dd60.
- Baur-Melnyk A, Birkenmaier C, Reiser MF: Lumbar disc arthroplasty: indications, biomechanics, types, and radiological criteria. Radiologe. 2006, 46: 768-78. 10.1007/s00117-006-1356-9.View ArticlePubMedGoogle Scholar
- Park CK, Ryu KS, Jee WH: egenerative changes of discs and facet joints in lumbar total disc replacements using ProDisc II: minimum two-year follow-up. Spine. 2008, 33: D1755-61. 10.1097/BRS.0b013e31817b8fed.View ArticleGoogle Scholar
- Bendo JA, Quirno M, Errico T, Spivak JM, Goldstein J: A comparison of two retroperitoneal surgical approaches for total disc arthroplasty of the lumbar spine. Spine. 2008, 33: 205-9. 10.1097/BRS.0b013e31816054d9.View ArticlePubMedGoogle Scholar
- Rauschmann MA, Thalgott J, Fogarty M, Nichlos M, Kleinszig G, Knap M, Kafchitsas K: Insertion of the artificial disc replacement: a cadaver study comparing the conventional surgical technique and the use of a navigation system. Spine. 2009, 34: 1110-5. 10.1097/BRS.0b013e31819e2235.View ArticlePubMedGoogle Scholar
- Petilon J, Hardenbrook M, Sukovich W: The effect of parallax on intraoperative positioning of the Charitß artificial disc. J Spinal Disord Tech. 2008, 21: 422-9. 10.1097/BSD.0b013e31815708a6.View ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.