- Case report
- Open Access
Skin avulsion injury during endotracheal tube extubation – case report of an unusual complication
© Yilmaz et al; licensee BioMed Central Ltd. 2008
- Received: 02 April 2008
- Accepted: 21 May 2008
- Published: 21 May 2008
We report a geriatric case with a full-thickness skin avulsion injury during extubation due to a tube securing tape used to fixate the endotracheal tube. The avulsed skin was sutured back to its original place. Based on this single geriatric patient, we recommend anesthesiologists/anesthetists and surgeons be aware of the potential risk of avulsing the skin with tape during a standard extubation procedure. This may especially occur in geriatric patients who have age related changes as decreased elasticity and resistance to shearing forces that predispose the skin to get traumatized easily.
- Endotracheal Tube
- Geriatric Patient
- Original Place
- Catgut Suture
- Accidental Extubations
Endotracheal tubes that are used for airway maintenance must be securely placed and fixated in order to avoid complications such as accidental extubations, tracheal injuries or swallowed endotracheal tube .
We report a 75-year-old patient in whom a full-thickness skin avulsion injury occurred during extubation due to securing tape used to fixate the endotracheal tube. In addition, plausable reasons of this rare complication are discussed.
Airway control is the first step of patient care requiring either short term or long term respiratory support, regardless of the primary pathology. Proper endotracheal intubation and safe fixation of the tube to prevent tube dislodgement are two critical steps for airway maintenance [2–6].
Several articles discussed the tube fixation methods in neonates and pediatric cases, because of their delicate skin structure to avoid complications [1, 3]. We believe special attention must also be paid in geriatric patients.
Cumulative effect of several factors seemed to have caused an avulsion skin injury in the presented case. One such factor was the aging of the skin. Aging is a process of loosing the properties of skin as well as resisting to the gravitational forces. At the histologic level, cutaneous aging is manifested by flattening of the dermal-epidermal junction, a decrease in the number of melanocytes and Langerhans cells, a reduction in the amount of glycosaminoglycan ground substance, progressive dropout of elastic fibers, and diminution in the total amount of collagen as well as the fraction of type III collagen. Superimposed on the inevitable changes outlined above are the effects of sun damage, which result in epidermal dysplasia and dermal elastosis, and dramatically exacerbate the deterioration inherent in the aging process. The histologic deterioration of the skin due to aging correlates with the clinical findings: thinning of the skin, decreased resistance to shearing forces, decreased elasticity .
Another important factor that may have contributed to this compliction was the presence of prior blunt injury to the right cheek. Additionally, the pulling force of the tape applied by the anesthesiologist also seemed to have played a significant role in the occurrence of this unfortunate injury.
We recommend anesthesiologists/anesthetists and surgeons be aware of the potential risk of avulsing the skin with tape during a standard extubation procedure. This may especially occur in geriatric patients who have age related changes such as decreased elasticity and resistance to shearing forces that predispose the skin to get traumatized easily, especially when a preexisting blunt trauma is present. We believe this complication could have been avoided by securing the tape to a non-traumatized skin areas, by less forceful and careful pull of the tube and the tape.
A written informed consent was obtained from the patient for publication of this case report and any accompanying images. A copy of the written consent is available for review by the Editor-in-Chief of this journal.
- Grammatikopoulos T, Devane S, Hannam S, Greenough A: Method of endotracheal tube fixation and iatrojenic ear deformities. Eur J Pediatr. 2003, 162: 41-3. 10.1007/s00431-002-1020-y.View ArticlePubMedGoogle Scholar
- Walker I, Lockie J: Basic tecniques for anaesthesia. Paediatric Anaesthesia. Edited by: Sumner E, Hatch DJ. 1999, London: Arnold, 165-210.Google Scholar
- Eskinder H: A new method of endotracheal tube fixation fot pediatric neurosurgical patients. Anesthesiology. 1993, 78: 618-9. 10.1097/00000542-199303000-00047.View ArticleGoogle Scholar
- Donlon WC, Truta MP, Hilt DM: A simple method of nasoendotracheal tube fixation. Ann Plast Surg. 1989, 23: 461-2. 10.1097/00000637-198911000-00017.View ArticlePubMedGoogle Scholar
- Mikawa K, Maekawa N, Goto R, Yaku H, Obara H: Transparent dressing is useful for the secure fixation of the endotraceal tube. Anesthesiology. 1993, 75: 618-9.Google Scholar
- Hansen RH, Remensynder JP: External fixation of endotracheal tubes while skin grafting severe burns of the face. Plast Reconstr Surg. 1978, 62: 628-9.View ArticlePubMedGoogle Scholar
- Thorne CHM, Aston SJ: Aesthetic surgery of the aging face. Grabb and Smith's Plastic Surgery. Edited by: Aston SJ, Beasley RB, Thorne CHM. 1997, Philedelphia-New York: Lippincott-Raven, 633-49.Google Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.